[Sara-Reports] SARA-KP Nightly Observing Report
Dr. Peter Mack
pmack at astronomical.com
Sat May 25 10:31:24 CDT 2019
I found the telescope displaying west horizon limit and 6h15m in the west this morning.
I tried to bring the telescope back to the zenith but it would not move (despite my under-the-hood commands). The DIO-96 was behaving erratically flashing warning signs. I rebooted the computer.
The display then read 1h26m in the west, and problem went away. Something is interfering with the
DIO-96 and I don’t know what it is.
From: sara-reports-bounces at tamuc.edu <sara-reports-bounces at tamuc.edu> On Behalf Of Veronique Petit
Sent: Friday, May 24, 2019 5:44 PM
To: sara-reports at tamuc.edu
Subject: [Sara-Reports] SARA-KP Nightly Observing Report
SARA-KP Nightly Observing Report
University of Delaware
vpetit at udel.edu <mailto:vpetit at udel.edu>
ARC CCD, Spectrograph
Equipment Failure (tracking/CCD)
Blazar monitoring and spectroscopy of magnetic massive stars
Nominal track rate, filter(s), ambient temp and focus:
Problems encountered, if any:
* We were connected to both computers through VPN on a mac. We lost the VPN connection to the telescope computer, and could not reconnect. The camera computer was still working fine, and Radmin on a PC was still working for the telescope computer.
* We were trying to do 30 minutes exposure with the spectrograph.The auto-guiding with the ASCOM camera was working very well, until the RA tracking (and the autoguiding) would just shut itself off. This happened on the 4 exposures we attempted, but not only when low in the south west -- for the last exposure, the telescope was pointing at HA = -2h, alt=70°.
Suggestions for improvements:
This email was built and sent using Visual Form Builder <https://urldefense.proofpoint.com/v2/url?u=http-3A__wordpress.org_extend_plugins_visual-2Dform-2Dbuilder_&d=DwMDaQ&c=oqyuZuih6ykib6aKiBq22_bich4AVfYGoLertJN0bEc&r=YOTRq01KpCKCeuVmK6YHaZpdkTJmABcEvt7TXktF7Cc&m=2Krcr9K3dIlW-P13aqUSRjrE1BsBul7oayXnlQV23Mo&s=8g8q-SVjwicXzijW1gU2kxTwvI3CY10SnQLap-cXNuo&e=> .
This email has been checked for viruses by AVG.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Sara-Reports